Hightop Biotech Coronavirus Antigen Rapid Test (1 Test)
The Coronavirus Antigen Rapid Test is intended for in vitro qualitative detection to SARS-CoV-2 antigen in human nasopharyngeal swab or oropharyngeal swab samples. For professional use.
Our product is in stock.
Discount price: €6.97 €5.57/test
The price includes 5% VAT.
SARS-CoV-2 Antigen Rapid Test (Immunochromatography)
Specimen type: nasopharyngeal swab or oropharyngeal swab
Kit size: 1 Test/Box
Manufacturer: Hightop Biotech Co., Ltd.
Main components of the Coronavirus Antigen Rapid Test Kit
Sample extraction tube
COVID-19 antigen rapid test sample requirement
Nasopharyngeal swab collection method
The operator holds the swab by the right hand and holds the head of the subject fixedly by left hand. Do not overexert to avoid traumatic hemorrhage. When the cusp of the swab touching the paries posterior of the pharyngonasal cavity, letting the swab remain in the place for a few seconds (about 3 seconds) and rotating the swab gently for one cycle,and then remove the swab slowly. Using the same swab, repeat this process for the other nostril to ensure that an adequate sample is collected from both nasal cavities.
Collection method of oropharyngeal swab
The head of the person to be collected is slightly tilted and his mouth is wide open, exposing the pharyngeal tonsils on both sides. Wipe the swab across the root of the tongue. Wipe the pharyngeal tonsils on both sides of the person to be collected back and forth with a little force for at least 3 times, and then wipe up and down the posterior pharyngeal wall for at least 3 times.
Coronavirus Antigen rapid test procedure
Instructions must be read entirely before taking the Covid-19 antigen rapid test. Leave the reagent and sample at room temperature for 30 minutes before use. Return to room temperature. Do not open the inner packing until it is ready. Use it as soon as possible after opening the inner packing.
Add 500μl (~20 drops) of sample extract to the 0.5 mark of the sampling tube, dip the swab after collecting the sample into the sample extract, make the sample extract fully permeate the swab, rotate and squeeze the swab 10 times, then pull out the swab, and take the stranded liquid as the sample to be tested.
Open the tear hole of the aluminum foil bag, take out the test card and lay it flat.
Apply 2-3 drops of the treated sample extract (60μl-80μl) vertically into the sample well of the test cassette.
The results are observed after 15 minutes and showed no clinical significance after 20 minutes.
The reagent is a disposable diagnostic reagent in vitro, which is only used for the detection of human nasopharyngeal swab, or oropharyngeal swab. The operation should be carried out strictly according to the instructions. Do not use expired and damaged products.
The strength of the quality control line does not mean the quality of the reagent, as long as its color is clear and visible, that means the reagent is effective.
The coronavirus antigen rapid test kit should be sealed and kept away from moisture. Reagents or samples stored at low temperature should be balanced to room temperature before they can be used.
Reagents should be used as soon as possible after removal from aluminum foil bags, so as to avoid exposure to air for too long and affecting test results due to dampness.
Do not use samples that have been placed for too long or contaminated.
Please operate in accordance with the laboratory testing procedures for infectious diseases. Waste after use should be treated in accordance with infectious substances and should not be discarded at will.
Incorrect operation may affect the accuracy of the results, such as insufficient sample mixing, insufficient amount, inaccurate detection time, etc.
Components in different batch should not be mixed.
There should be appropriate biosafety assurance procedures for those substances containing and suspected sources of infection.
The following are relevant considerations:
- Handle samples and reagents with gloves;
- Do not suck samples with your mouth;
- Do not smoke, eat, drink, cosmetic or handle contact lenses while handling these items;
- Disinfect the spilled sample or reagent with disinfectant;
- Disinfect and treat all samples, reagents and potential pollutants in accordance with relevant local regulations;
- Each component of the reagent remains stable until the expiry date under proper handling and storage conditions. Do not use the expired reagent kit.